Theparameter can be numeric or a string. General Use Programming clear selected. For example, you may program the dc source to generate an interrupt assert SRQ when an eventsuch as a current limit occurs. Are the E series power supplies two quadrant? Asshown in figure , the group consists of the same type of registers as the Status Operation group. WTG is the logical-or of both transient and acquire sequences. The data may be numerical or characterstring.
|Date Added:||4 June 2016|
|File Size:||42.44 Mb|
|Operating Systems:||Windows NT/2000/XP/2003/2003/7/8/10 MacOS 10/X|
|Price:||Free* [*Free Regsitration Required]|
COMPatibility Commands continued DescriptionThese commands determine the conditions that will set bits in thefault register, allowing the operator to define the conditions that willbe reported as fault Fault conditions can agileng enabled by sendingthe decimal equivalent of the total bit weight of all conditions to beenabled. The Configuration VIs- a collection of software routines that configure the instrument to perform the operation you want.
This section describes the output trigger system. If the password is notentered or is incorrect, an error is generated and the calibration mode remains disabled. January 10, 5: For example, to set the output voltage to 25volts, use: It normally serves as an output. Generally, you need to call the Close VI only once at the end of an application or when you finish communication with an instrument.
D – Example ProgramsExample 2. Language Dictionary – 6632g CommandsMeasurement commands consist of measure and sense commands.
DATAThis command enters a calibration value that you agilebt by reading an external meter. Two types of measurement commands areavailable: LEVel for the value being entered. The Status VIs obtain the current status of the instrument or the status of pending operations. The calibration mode must be enabled before thewill accept any other calibration commands.
Table defines the status bits. The dc source responds to the following commands: December 20, 5: The following table lists the errors that are associated with SCPI syntax errors and interface problems.
This disables all frontpanel keys except 66332b Local key. How to set NB in order to show 0. Figure shows the status register structure of the dc source. The Hanning4window multiplies each point in the measurement sample by the function cos.
Page 61 Language Dictionary – 4Status CommandsStatus commands program the dc source status registers. With this command,the trigger system needs to be initialized only once at the start of the acquisition period.
The fastest way to get up and running with your instrument driver is to open one of the pre-built examples. PROTection command asexplained in chapter 4.
DISPlayThis command turns the front panel display on or off.
Common commands also perform system functions. The Utility VIs- perform a variety of operations that are auxiliary to the most often used instrument driver VIs. Back to Top 2. November 20, 5: The output mode switch is used to connect ordisconnect the the output capacitor located inside the unit.